LTV580 portable color doppler ultrasound machine
abdomen,maternity, urology, superficial organization, peripheral vascular, small organ, cavity, and other interventional screening and therapeutic use.
2.1full-digital color ultrasound diagnostic system, the system has the ability to upgrade
2.2 Frequency Range :2.0-14MHz
2.3 Probe types: convex array, linear array, heart and cavity probes
2.4 Display: B / M, B / D, B / CD, B / D / CD may be the same screen display, M type and scope of PWD icon size adjustable
2.5Frequency Probe Specifications:
2.5.1 Abdomen convex array probe: 2.0 ~ 5.0MHz electronic convex array broadband probe frequency, the frequency range 2 - 5MHz, 3 selectable frequencies Bandwidth
2.5.2 E-broadband frequency linear array probe, the frequency range 5 - 14MHz, scan angle: 60 degrees
2.5.3 Intracavity frequency convex array broadband probe, the frequency range 4 - 8MHz, scanning angle ≥ 135 °
2.5.4 Optional pediatric cardiac probe, transesophageal probe, laparoscopic probe, tiny protruding array probe, intraoperative probes.
2.6 probe array element: ≥ 128
2.7 Image Mode: two-dimensional B-, M-type, pulsed PW , color CFM / power Doppler and direction of the Energy PDM
2.8 B / D use either: B / PWD, B / CD / PWD three simultaneous display
2.9 Digital full anatomical M-imaging techniques, the sampling line can be anywhere in the 360-degree range as the center of arbitrary sampling
2.10 Organization of second harmonic imaging, harmonic function ≥ 2 groups
2.11 B / M / CD can be adjusted independently
2.12 Resolution: Lateral ≤ 2mm, longitudinal ≤ 1mm
2.13 detecting depth: ≥ 240mm
2.14 system dynamic range ≥ 140dB
2.15 15cm deep, full-view scans, anatomical M-frame rate ≥ 120 / s (for pictures)
2.16 Image playback video: frame by frame, continuous playback of ≥ 300 frame
2.17 Focus: Emission ≥ 8 segment focus, to receive: continuous dynamic variable aperture, dynamic apodization digital focusing
2.18 Scan Line: Every frame linear density ≥ 400 Ultrasonic Line
2.19 Measurement and Analysis
2.19.1General measurement: distance, perimeter, area, slope, heart rate, pressure, residual urine, RI, PI, flow rate, etc.
2.19.2 Gynecology and obstetrics special measuring software: fetal gestational age ,EDC, estimation fetal weight, Take average measured multiple times,S/D ratio, etc
2.20 Character Tags: shows the date, time, patient's name, user name, etc., custom note table, probe, frequency and body marked, with a puncture and guide lines
2.21 keyboard operation: Sino-British operation interface
2.22 position markers: ≥ 30 Zhong with the location of the probe position markers
2.23 Color Doppler display modes: speed dispersion shows that the energy shows that dispersion Show
2.24 display position adjustment: linear array scanning range of interest: -20 ° - +20 °
2.25 Doppler flow velocity: The maximum blood flow velocity measurements: PWD ≥ 6m / s; the highest single measurement of a continuous Doppler velocity ≥ 10 m / s minimum flow velocity measurements: PWD ≥ 10mm / s
2.26 width and location of sample volume adjustment :0.5 - 20mm adjustment classification
2.27 with a Doppler angle correction function for sampling and then
2.28 Display: ≥ 15-inch high-resolution progressive-scan LCD displays.
2.29 Probe Interface:2 units
2.30 picture archiving and management
2.30.1Dynamic and static image real-time hard drive storage capabilities, the host built-in hard disk ≥ 160G.
2.30.2Built-in DVD drive with a recording can be carved directly to the medical records of the CD-ROM to save.
2.30.3 Playback of the spectral image can also be synchronized audible sounds
2.30.4There was a picture online clipboard functions: real-time scan, only one button operation, can be dynamic and static ultrasound images are stored in the screen side of the clipboard, you can always transfer out of contrast observation.
2.30.5 The original data acquisition and processing capabilities, can playback dynamic and static image post-processing capabilities, and can convert directly to the image avi, tif, Bmp, and other common format computer
2.31 Input / Output Signal Interface: PAL-D, USB, RS-232, VCR, RGB, USB,, VGA port, etc. 2.32 Supply Voltage: AC
Where are our customers?
We have sold our products to more than 50 countries ,like The Democratic Republic of Congo, Saudi Arabia, Nigeria, Zambia, Angola, Egypt, Kenya, Ghana, Paraguay, Nepal and Myanmar and so on.
|Your satisfaction is our duty! It will be our great honor to serve you.|
Our main products include X-ray machine, B ultrasound scanners, ECG, patient monitor, ICU equipment, Lab equipment, and other related medical disposables.
We have exported our products to more than 50 countries and built long term partnerships with clients.
We do hope we can offer you more help in medical field and build the long term business relationship with you.
|Delivery Details||1.We ship goods by UPS/DHL/FEDEX/ TNT express takes 3~5 days to arrive , which is door to door, you no need go out can get LTV580 portable color doppler ultrasound machine.|
2.We can also help you ship by air cargo, it takes about 2-7 days to arrive, while you need to go to Air port to take LTV580 portable color doppler ultrasound machine.
3.If goods are large quantity and you need not so urgently ,you can choose by sea to save some freight cost which takes you about one month to get LTV580 portable color doppler ultrasound machine.
No matter what delivery way ,depending on your actual requirements...
Q:What’s the minimum order quantity (MOQ)?
A:For most of our medical products, even order for only one unit is warmly welcomed.
Q: Can you do OEM/ private label?
A: Of course we can do OEM/private label for you.
Q: What’s your delivery time?
A: Generally it is 10 days if the goods are in stock.
or it is 15-20 days if the goods are not in stock, it is according to quantity.
Q: How to ship the order?
A: Please inform us your instruction, by sea, by air or by express, any way is ok for us. We have very professional forwarder to provide the best shipping cost, service and guarantee.
Q: What is your terms of payment ?
A: We accept T/T ,LC,Western Union ,Escrow ,Paypal and more.Please suggest your preferred payment method.