64 Slice Spiral CT
1, Technical specifications and parameters
1.1.1 Frame size: 76cm, angle: ±30 degrees, can be remote control operation into / out, rise / fall table and gantry tilt
1.1.2 Slip ring type: contactless high speed data link slip ring
1.1.3 Detector type: GOS solid detector
1.1.4 Detector row: 32
2.1 X-ray system
2.1.1 Tube anode heat capacity: 8.0MHU
2.1.2 Maximum anode heat dissipation rate of tube: 931 KHU/min
2.1.3 Small focus: 0.6 x 1.2 mm
2.1.4 Large focus: 1.1 x 1.2 mm
2.1.5 Focus to the scanning field and other central distance: 570mm
2.1.6 Minimum tube voltage: 80kV
2.1.7 Maximum tube voltage: 140kV
2.1.8 High voltage generator power: 80kW
2.1.9 Minimum current: 10mA
2.1.10 Maximum current: 667mA
2.2 Patient Table
2.2.1 Maximum scan range: 1850mm
2.2. The highest vertical movement of the table is 990mm,
the lowest vertical movement of the table is 425mm
2.2.3 Maximum table longitudinal speed: 150 mm/s; minimum longitudinal speed: 5 mm/s
2.2.4 The table longitudinal movement accuracy: ±0.25 mm
2.2.5 The maximum patient table load: 250kg
2.3 Console and computer system
2.3.1 The main console
18.104.22.168 Master computer: HP Z820
22.214.171.124 Computer memory: 32GB
126.96.36.199 Host hard drive: 7TB
188.8.131.52 CD-ROM disc recorder system (DICOM compatibility)
184.108.40.206 Color LCD flat-panel monitors flicker 1 unit (progressive)
220.127.116.11 Dtandard DICOM3.0 interface: standard configuration, with the following functions:
Send / receive
Query / retrieval
Basic print function
Network interface (RIS / HIS)
18.104.22.168 Laser camera interface
22.214.171.124 Remote diagnosis system and remote diagnosis platform
126.96.36.199 Virtual photo type setting function
188.8.131.52 Console color 3D display
2.3.2 Advanced workstation
184.108.40.206 Offers a package of clinical image processing software, such as MPR, VR, SSD, the maximum and minimum software package Projection; standard configuration volume measurement software.
2.4 Scan parameters
2.4.1 Scan time: 0.39s, 0.5s, 0.6s, 0.7s, 0.8s, 1.0s and 2.0 sec
2.4.2 Imaging speed (number of scan per scan): 64slice/loop
2.4.3 Scan Slice thickness: 0.625mm, 1.25mm, 2.5mm, 3.75mm, 5mm, 7.50mm, and 10mm
2.4.4 Minimum reconstruction display field (FOV): 50mm
2.4.5 Maximum reconstruction display field (FOV): 500mm
2.4.6 Image reconstruction matrix: 512 * 512, 1024*1024
2.4.7 Single scan time: 100 sec
2.4.8 Maximum scan pitch (Factor Pitch): 1.75:1
2.4.9 Minimum Factor (Pitch): 0.15:1
2.4.10 Positioning piece length: 1950 mm
2.5 Image quality
2.5.1 Spatial resolution (X, Y axis plane):20 lp/cm 0% MTF
2.5.2 Spatial resolution (Z axis):20 lp/cm 0% MTF
2.5.3 Density resolution: (indicating the scan conditions): @0.30% 3mm
2.5.4 CTDIvol: Head 16.6mGy/100mAs Body; 8.6mGy/100mAs
Where are our customers?
We have sold our products to more than 50 countries ,like The Democratic Republic of Congo, Saudi Arabia, Nigeria, Zambia, Angola, Egypt, Kenya, Ghana, Paraguay, Nepal and Myanmar and so on.
|Your satisfaction is our duty! It will be our great honor to serve you.|
Our main products include X-ray machine, B ultrasound scanners, ECG, patient monitor, ICU equipment, Lab equipment, and other related medical disposables.
We have exported our products to more than 50 countries and built long term partnerships with clients.
We do hope we can offer you more help in medical field and build the long term business relationship with you.
|Delivery Details||1.We ship goods by UPS/DHL/FEDEX/ TNT express takes 3~5 days to arrive , which is door to door, you no need go out can get 64 Slice Spiral CT.|
2.We can also help you ship by air cargo, it takes about 2-7 days to arrive, while you need to go to Air port to take 64 Slice Spiral CT.
3.If goods are large quantity and you need not so urgently ,you can choose by sea to save some freight cost which takes you about one month to get 64 Slice Spiral CT.
No matter what delivery way ,depending on your actual requirements...
Q:What’s the minimum order quantity (MOQ)?
A:For most of our medical products, even order for only one unit is warmly welcomed.
Q: Can you do OEM/ private label?
A: Of course we can do OEM/private label for you.
Q: What’s your delivery time?
A: Generally it is 10 days if the goods are in stock.
or it is 15-20 days if the goods are not in stock, it is according to quantity.
Q: How to ship the order?
A: Please inform us your instruction, by sea, by air or by express, any way is ok for us. We have very professional forwarder to provide the best shipping cost, service and guarantee.
Q: What is your terms of payment ?
A: We accept T/T ,LC,Western Union ,Escrow ,Paypal and more.Please suggest your preferred payment method.